The technique description herein is made available to the healthcare professional .. to surgeon preference and/or fracture pattern. IMHS CP Nail Extractor. Surgical Technique. International Version. Page 2. Page 3. Nota Bene. The technique description herein is made available to the healthcare professional to illustrate the . enhanced sliding and compression of the IMHS™ IMHS™ CP Nail. Patient positioned with operative extremity adducted for intramedullary nailing. Distal interlocking of these longer nails requires a freehand technique. models of each of these devices have been introduced—Gamma III and IMHS CP.
|Published (Last):||23 December 2018|
|PDF File Size:||12.76 Mb|
|ePub File Size:||14.51 Mb|
|Price:||Free* [*Free Regsitration Required]|
No Multipurpose Wrench Cat. Description mm x 70mm mm x 75mm mm x 80mm mm x 85mm mm x 90mm mm x 95mm mm x mm mm x mm mm x mm mm x mm mm x mm mm x mm mm x surgicall mm x mm mm x mm IMHS CP Captured Locking Screws Cat No. The information contained in this document is intended for healthcare professionals only.
No Mini Connector Cat.
Attach the T-Handle to the Hex Driver and insert the screw. Other new features A more anatomical 2. Ream until the proximal femur is prepared to accept the proximal portion of the implant.
IMHS CP Clinically Proven Intramedullary Hip Screw Surgical Technique – PDF
Description Nail Cap Sample L x 10mm x The nail is properly placed in the AP plane when the Guide Pin is seen on the X-Ray to be parallel to the neck and center in the femoral head. Instruments and implants approved by.
No mm Guide Pin Cat. Our strategy has been to know.
IMHS CP Clinically Proven Intramedullary Hip Screw Surgical Technique
Take the screw length measurement from the calibrations on the Drill Bit as it relates to the end of the Screw Length Half Sleeve. HN Compression Dial Cat. Screw length may also be determined using the Screw Depth Gauge.
Begin reaming with the 9. No Cannulated Awl Cat. Titanium Trochanteric Fixation Nail System. Attach the appropriate length screw to the Short Hexdriver, connect the T-Handle and insert imus screw until it is seated.
Adolescent Lateral Femoral Nail. No Screw Depth Gauge Cat. In the final analysis, the preferred treatment is that which addresses the needs of the specific patient. Connect this assembly to the Subtrochanteric Lag Screw. Cerclage solutions for general surgery. If not satisfied with depth, continue reaming and confirm depth with X-Ray. No This illustration depicts an older design where the side plates have now been removed, as well as the mating holes for the Anteversion Locking Guide.
Table of Contents More information. Additional reaming may be required, or, in some instances, a smaller implant diameter may be selected.
Image intensifier control Warning. No Guide Bolt Cat. Example If the measured length was 95mm and a mm screw was selected, the Compression Dial on the Insertion Wrench must be set to 5. To help ensure the accuracy of the drill guide, it is helpful to tighten the Guide Bolt with the Tissue Protector and Lag Screw Reamer in place. While this is a stopped drill, care should be taken to ensure only the lateral cortex is over drilled.
It is inserted into the Telescoping Reamer Sleeve until it abuts the end of the sleeve. For intramedullary fixation of proximal femur fractures. Slide the Ruler over the proximal end of the Ball-Tip Guide Rod and advance the open end of the Ruler to where the proximal portion of the implant will be seated, just below the tip of the greater trochanter.
Parallel T-Handle position limited sliding Perpendicular T-Handle position more sliding This illustration depicts an older design where the side plates have now been removed, as well as the mating holes for the Anteversion Locking Guide.